Enrique used the setup made by Tibi to characterize and optimize the tape transducer and recorded the first comprehensive data.
The main goal of this experiment was to find the optimal size of the gap (distance between the mirror and the fibers).
The one in 3 out prototype with round glass fiber tip was used. The setup consisted of a green laser coupled to the 125/62.5 MM glass fiber (the in fiber) using a lens. The fiber was hold in place using the reusable ST connector. The one in 3 out prototype (fiber assembly) was prepared by Tibi before the experiment: PMMA fibers were polished and the glass fiber tip was cleaned with alcohol. The tip of the fiber on the side of laser was freshly cleaved.
Setup used during this experiment
Preliminary data obtained
We are looking for a linear relation between angle (mirror tilt) and gap size. The preliminary data shows that being too far from, or too close to the mirror is not good. There is a sweet spot somewhere in the middle. Open the spreadsheet used for characterization and optimization of the tape transducer. Next week we'll continue exploring this same prototype and evaluate its sensitivity. Tibi needs to modify the LabView program in order to allow faster acquisition and better statistics.
See the Tape Sensor optimization and characterization doc.